Following the discovery that wind monitor orientation discrepancies were adversely affecting wind direction data in Wind speed and direction on lakes on-buoy, NEON has field-verified current wind monitor orientations at the buoy sites and updated the algorithm to account for the wind monitor orientation and approximate uncertainty atop the buoy mast.
neonUtilities 2.1.0 is now available on CRAN! This new version adds new options for handling Release data, improves error messaging, and fixes bugs in flux footprints and microbe community composition. Update is highly recommended.
This is an update to the previous post concerning external lab surface water and groundwater nutrient data. Currently, sub-samples for nutrient analysis are being frozen and stored for future analysis with lower MDLs. The newly contracted external laboratory will begin receiving samples June 14, 2021 and all stored samples will be shipped by December 31, 2021. Data will be published approximately 90 days following sample receipt.
Surface water microbe cell count (DP1.20138.001) laboratory data for all aquatic sites from 9/2017 to 9/2020 were found to have incorrect calculations populating rawMicrobialAbundance data field. Data were recalculated, edited, and published to the data portal on April 5, 2021.
Macroinvertebrate metabarcoding (DP1.20126.001) and Zooplankton metabarcoding (DP1.20221.001) bioinformatics data in the tables inv_metabarcodeTaxonomy and zoo_metabarcodeTaxonomy tables are missing or have incorrectly labeled records for COI-F230 for samples collected in 2018 and 2019.
Are you interested in investigating spatial and spectral relationships between high resolution airborne hyperspectral data and satellite observations? Learn more about our September 2020 data collection campaign at CPER in conjunction with Landsat7, Landsat8, and Sentinal 2A.
Macroinvertebrate collection (DP1.20120.001) field data in the table inv_fieldData for all aquatic sites from 2014 through 2019 are duplicated on the NEON data portal due to an error on data transition for this table. Corrections have been made, but will not appear on the portal until the 2022 data release.
This is an update to the previous post concerning wind direction data from the buoys. NEON has field-verified current wind monitor orientations at the buoy sites, and is taking multiple corrective measures.
AIS field maintenance and calibration data are now available for the following data products for provisional data (primarily data collected July 1st, 2020 and after). Read on to see which data products provide these tables.
The NEON microbial community composition data products have been re-processed and are available on the NEON data portal either as provisional data (data from December 2014 and earlier) or in the initial NEON data release (data from 2015 through October 2018) that is available now.
AOP has updated the file name, format and metadata for the Spectrometer Ortho-rectified at-sensor radiance data product (DP1.30008.001). The compression of the data has improved, and updates to end users' code may be needed.
This is an update to the previous post concerning wind direction data on the buoys. NEON has field-verified wind monitor orientation discrepancies at most buoy sites, which adversely affect wind direction data in Wind speed and direction on lakes on-buoy (DP1.20059.001). Buoy wind direction data will be corrected with measured offsets and/or quality flagged following further review over the next 6 months.
neonUtilities 2.0.0 is now available on CRAN! This new version contains critical updates for working with the NEON Data Release and associated API and Portal changes. An update to 2.0.0 is required for working with any newly downloaded data.
Based on an analysis of nutrient data, NEON has determined that the method detection limits (MDLs) required of the external laboratory for surface water and groundwater nutrient analyses are too high for many sites. New analyses of frozen samples will take place in the future.
Surface water microbe cell count (DP1.20138.001) laboratory data for all aquatic sites from 9/2017 to 9/2020 were found to have incorrect calculations populating rawMicrobialAbundance data field. Data have been removed from the data portal while revisions are made and will be republished by February.
Due to a miscommunication, samples analyzed for carbon (C) and nitrogen (N) concentrations and stable isotopes were not previously re-dried prior to weighing and analysis at the external laboratory. Affected data have been flagged and a new protocol was implemented.
To reduce uncertaintly in surface and groundwater elevation data, permanent benchmarks have been installed at most aquatic sites to serve as controlled location reference points. Updated elevations from the associated site resurveys are available on the data portal. Surface and groundwater elevation data products have been reprocessed, and, the corrected data will be available via the data portal by January 18, 2020.
We have discovered an error in the larval tick count data collected in 2016-2017 in Ticks sampled using drag cloths (DP1.10093.001) that affects all sites. Samples from the tick taxonomy laboratory had a count of 1 larva in the taxonomy table no matter how many individuals were present in the sample. The data will be rectified within 2-3 weeks.
The NEON microbial community composition data products have been re-processed and will be available on the NEON data portal either as provisional data (2014-12 and earlier data, 2018-10 and later data) or in the initial NEON data release (2015 through 10-2018 data) in January 2021.
The sunlit plant foliage major, minor, and trace element data (DP1.10026.001, cfc_elements table) collected in 2017 and 2018 were investigated and deemed untrustworthy. Sample major, minor, and trace elements were re-measured using foliar material from the NEON Biorepository. Following review, these new data were posted to the Data Portal on 2020-10-29.
All NEON wind direction data at all sites under-report average wind direction frequency in the ~360 to ~5 degree range. This includes wind direction data in both 2D Wind speed and Direction data products: DP1.00001.001 and DP1.20059.001.
At all sites with profiling water quality sondes on buoys, the Water quality data product (DP1.20288.001) reports the depth of the multisonde as it profiles through the water column. Data users should be aware that sensor depth is approximate, as the multisonde’s pressure transducer is not vented to the atmosphere. Therefore, barometric pressure fluctuations following calibration will cause a sensor depth discrepancy.
Errors were discovered in NEON.DOC.004931 (version A), the Algorithm Theoretical Basis Document (ATBD) for the Water quality (DP1.20288.001) data product. These errors impact the temperature and absorbance corrections applied to fluorescent dissolved organic matter (fDOM) data.
All sites with buoys (SUGG, BARC, FLNT, CRAM, LIRO, TOMB, BLWA, PRPO, PRLA, TOOK), need to be verified for appropriate wind direction data from the Wind speed and direction on lakes on-buoy data product (DP1.20059.001).
The Airborne Remote Sensing Data Quality Technical Working Group recently made a series of recommendations for changes to the NEON AOP data product catalog. These changes included data product name changes, as well as suspension of a subset of AOP products.